Central Office Address:
No.104,Andarzgoo Blvd,Qeytarieh area,Tehran,Iran
No, abadis canisters do not need to be replaced after each use.
The suction bag is made of polyethylene (PE) and the canisters are made of polycarbonate.
These filters are made of nanoparticles and polymer and composite materials and makhazen tebi abadis Company is introduced as the first manufacturer of these filters in Iran.
Abadis Medical canisters Company is the first manufacturer of these filters in Iran.
Absorbent powder in Abadis suction bags causes jelly of liquid materials inside the bags, which makes it easy to carry and also in case of overflow or torn bag, causes better collection of liquids inside the bag.
Suction bags can be replaced after full filling and be sure not to move the suction bags of different patients together to better control the infection.
1) Don’t use a disposable suction bag that fits the size of the canister.
2) suction Bags shouldn’t be replaced at least once every 24 hours if used.
3) Use the suction bags more than once.
This product will be out of warranty if the Abadis disposable suction bag isn’t used properly in the above cases.
1) Use disinfectants and water in the ratio of 1 to 10 to wash the suction canisters.
2) Roll the suction canisters for 10 minutes in the solution mentioned above; Be careful not to put the canisters in solution for more than the mentioned time because it will damage the canisters.
3)after that place the canister on a dry towel and allow the canisters to dry completely.
4)Abadis suction canisters can be washed in autoclave up to 120 ° C.
5)The canister must be replaced if there are any cracks or fractures.
How to wash the canisters is only acceptable as above, and if the washing is not done in this way, it will be out of warranty.
* Keep in mind that Abadis suction canisters only need to be washed if there is contamination.
When the paper inside the antibacterial filter changes color, the filter must be replaced, and if the filter is changed.
flow stopper filter should be replaced when fluids penetrate the filter surface or the suction pressure drops, and it is best to replace the circuit breaker filter every month for proper operation.
Yes, wall and portable brackets in both aluminum and plastic can be washed up to a maximum temperature of 95 degrees.
Abadis trolley is out of warranty in case of severe impact.
Abadis suction canister will be out of warranty if it is damaged.
Abadis regulator will be out of warranty if it is hit and the device is damaged.
Abadis disposable suction liner bag if have any problems can be replaced up to one year
Abadis disposable suction canister if have any problems can be replaced up to one year
Abadis trolleys if have any problems can be replaced up to 6 months After that, the repair is possible at a cost.
Abadis wall bracket if have any problems can be replaced up to 3 year
Abadis rail bracket if have any problems can be replaced up to 3 year
Abadis regulator if have any problems can be replaced up to 6 months After that, the repair is possible at a cost.
wall suction if have any problems can be replaced up to 6 months After that, the repair is possible at a cost.
In each carton, there are 40 disposable suction bags for every 3 sizes
Abadis disposable suction bags are produced in three sizes of one, two and three liters.
20 foot container
40 foot container
You can make your request by filling out the contact form with us or contact the installation unit of Abadis Medical Tanks Company.
Yes, you can see the following image:
up to four canisters can be connected in a series.